Your shopping cart is empty!
|Calibrate the 11063 by immersing in a cup of ice water.
Press the CAL button and allow it to cycle.
Thermometer is calibrated!
|Digital Probe Thermometer Auto-Calibration Procedure|
a. Take readings unti l a stable set of readings has been recorded.
If “Err 1” appears on the LCD display, fi rst reading is greater than 35.6°F (2°C). To clear, shut off thermometer and start again.
|If “Err 2” appears on the LCD display, the average reading was outside the +/- 0.54°F (0.3°C) tolerance range. To clear, shut off thermometer and restart.|
|FlashCheck® Jumbo Display Auto-Cal Needle Probe Thermometer|
|This unique, next-generation, Needle Probe Thermometer is engineered to set the industry standard for accuracy, durability and readability. It is designed and constructed under exacting standards to meet and exceed specifications required for commercial and professional uses.
|Measurement Range||-40°F to 311°F (-40°C to 155°C)|
|Accuracy||1.0°F (-4°F to 113°F) ±0.5°C (-20°C to 45°C); otherwise ±1.8°F (±1.0°C)|
|Response Time||Less than 5 seconds (0°C to 25°C/32°F to 77°F)|
|Resolution||0.1°(F or C)|
|Display Size/Update||1.5” x 0.5” (38mm x 12.7mm)/Every 2 Seconds|
|Probe Length and Tip Diameter||Stainless steel probe, length 3.90 +/- 0.004 in.
(99 +/- 0.1mm); 14 Gauge - Needle probe 0.067 in. (1.7mm) dia.
|Factory Calibration||NIST traceable calibration certified|
|Compliance Certificate||Manufacturers certificate of compliance available from DeltaTRAK, NIST Traceable, NSF, CE|