Your shopping cart is empty!
|DISPLAY AND TEMPERATURE CONTROL|
A temperature sensor connected to the generator is applied against the outer glass plate of the sequencer. A temperature limit (0 ° to 99 ° C) can be set instead of a power limit. During the migration, when the preset limit value is reached, the charged voltage automatically decreases until the temperature of the glass plate (on which the measurement is carried out) is less than the preset limit.
|Range / Resolution||
10-1000V / 10V
|Minimum values Regulated||10V, 1 mA, 1 watt|
|Range Display / Resolution||0-3000V / 10V
0-300mA / 1mA
0-300watts / 1watt
0-99 ° C / 1 ° C
1-99999V-hour / 1V-hour
1-9999min. / 1 min.
|Minimum flow||10V, 15μA, 0.3W|
|Fault Detection||Storp and audible alarm|
|Current range with program # 28||0-600mA|
|Dimensions (WxDxH)||27 x 34 x 11|
|119050||Power supply for protein electrophoresis pattern PS1006P|