- Nous contacter
If you’re surveying buildings, inspecting a home, drying a building after a flood, or testing concrete moisture for a flooring application, the MMS2 is the solution for you.
Four-In-One Moisture Measurement System
|Measures moisture in wood and wood floors, drywall, concrete and concrete block, stucco, plaster, masonry and other building materials.
Measure (pin mode) diagnoses the extent of moisture intrusion for damage assessment and monitor drying out of building structures.
|Search (non-invasive mode) behind ceramic tile, fine finishes, water stains, tile and vinyl floor coverings, wood, drywall, plaster, masonry, concrete and concrete block.
|Hygrometry measures or monitors buildings for adequate ventilation affecting indoor air quality and moisture problems.
|Surface Temperature (IR) checks surface temperature, utilizing laser pointer, calculates proximity to dewpoint.|
When concise and accurate environmental readings need to be reported, the MMS Plus gets the job done.The ability to record readings instantly at the push of a button, and to continuously log when left on site, makes this instrument ideal for many applications including:
Humidity Probe Options
|The MMS Plus may be used with three styles of interchangeable humidity probe, the Hygrostick, the Quikstick and the Quikstick ST. The Hygrostick (grey POL4750) can be used for high moisture applications such as concrete measurement. Quikstick (black POL8750) is a general purpose, fast-responding full range sensor. Quikstick ST POL78751, standard with all MMS2 kits and with the same performance as the standard Quikstick. A Quikstick ST can remain connected to the MMS 2 while using the pins.|
|Hygrostick part number POL4750, for high moisture applications.|
MMS2 can be used for measuring equilibrium relative humidity directly in concrete slabs. Protimeter pioneered this more accurate method, which includes drilling holes in the concrete, inserting a humidity sleeve and measuring the equilibrium humidity in the concrete. If excessive moisture is found, simply replace the sleeve cap for future retesting. This method also permits normal construction without disturbing the test surface.
A small hole is drilled in the concrete. Next, a humidity sleeve is inserted and capped flush with the floor. The relative humidity of the air in the test hole is now at the same moisture level as the concrete around it. Flooring product manufacturers normally recommend RH readings between 75% and 85% depending on the permeability of the product being installed.
Readings from multiple Hygrosticks can be taken and recorded with ease. Humidity readings can be taken with the use of humidity sleeves or humidity box. Hygrosticks, not Humisticks, should be used for this test.
|Gross Weight||10 oz (283 gms)|
|Dimensions (LxWxH)||7.5 in x 3.7 in x 2.2 in (190.5 mm x 94 mm x 56 mm)|
|Maximum Needle Depth||0.4 in (10 mm)|
|Batteries (Included)||9 V|
|Warranty||24 months on mechanical or manufacturing defects. Does not include
wearing part or accessories
|Moisture Measurement Range||Pin (% WME) 8% to 99, readings over 30% are relative
Non-invasive (RF) up to 3/4 in (19 mm) deep
60 to 1000 (relative)
|IR Surface Temperature
|15°F to 120°F (-10°C to 50°C)|
|IR Based – With 12:1 (D:S) Ratio Range -4°F to 176°F (-20°C to 80°C) Accuracy ±3.6°F (±2°C)|
|Hygrostick Data (Nominal)||30% to 40% RH (±3% RH) at 68ºF (20ºC). 41% to 98% RH (±2%) at 68ºF (20ºC). Range 32°F to 122°F (0°C to 50°C) ±0.6°F (±0.3°C)|
|Quikstick and Quikstick ST Data (Nominal)||0% to 10% RH, ±3% RH at 68°F (20°C), 10% to 90% RH, ±2% RH at 68°F (20°C), 90% to 100% RH, ±3% RH at 68°F (20°C), Range 32°F to 122°F (0°C to 50°C) ±0.6°F (±0.3°C) Nominal response 30% to 90% and back to 30% RH in 45 seconds @ 68°F (20°C)|
|Data Storage||Store up to 8,000 results with date and time stamp from all instrument functions with the push of a button. Store 6100 results with automatic logging.|
in Hard Case
in hard case
in hard case
|Hygro / quick
|Canva Carry Pouch||Pou5800-01||Yes||Yes||No||No||No|
|Hard Carry Case||BLD5910||No||No||Yes||Yes||Yes|